HS-300 Textile and Leather Softness Abrasion Testing Machine
|1-2 3-4 5+|
|FOB Unit Price:||US $2,000 US $1,800 US $1,600|
|Purchase Qty. (Pieces)||FOB Unit Price|
|Payment Terms:||L/C, T/T|
- Model NO.: HS-300
- Maxcapacity: <1000KN
- Load Way: Mechanical Load
- Display: Digital
- Weight: 0-100Kg
- Trademark: HS
- Specification: N/A
- Type: Leather Test Machine
- Accuracy Grade: 0.5
- Loading Method: Dynamic Load
- Control: Manual Control
- Power Source: AC220V
- Transport Package: Wooden Case
- Origin: China Dongguan
This machine is designed based on IULTCS.IUP/36. It is characterized by accuracy, beautiful appearance, simple operation, easy maintenance and convenient portability.
This tester is used to ensure that the leathers in the same pack are of a similar softness or assess that the softness of every part of single leather is uniform.
35mm, 25mm, 20mm optional
530±10g or 500g
0.1-10mm(Practice scale 0.1-7.6mm)
Getting Start Guide
The tester is durable with high precision. It's used to test the softness index of leather. It is widely used in footwear, furniture and garment area.
A small internal weight is exerted to the probe, then put down the press arm with a certain speed to keep it naturally pressed against the leather specimen which fixed on the test seat.
The leather softness index is read through the dial of press arm.
3. Product Standard
The instrument is complied to standard IULTCS, IUP/36.
1. Instructment structure
A. Dial (0.1~10mm)
B. Press handle
C. Release Button of Press Arm
D. Upper Arm
E. Specimen Clip
Ring: dia 20mm*1pc
Pointer zeroing disc *1pc
4. Instrument Installation
Environment: room temperature
Please put the instrument on horizontal desktop.
5.1. Sample Preparation:
Customer prepares the sample itself or according to the specification of IULTCS.
5.2.1. Zero the dial A with pointer zeroing disc;
5.2.2. Press the press handle B and at the same time press the release button C to release the upper arm D, then the upper arm will rebound.
5.2.3. Put the pointer zeroing disc on the test seat, press B until the upper arm is locked and at the same time you can hear the lock sound, then move away the hand.
After that, the pointer of dial A should point at "0", if not, turn the dial to keep the pointer to "0".
5.3. Test Process:
5.3.1. Take away the pointer zeroing disc according to the process of 5.2.2, and put the leather specimen on the test seat. Press down the press handle to make the probe retract totally, then press the upper arm D until the specimen is cliped and the upper arm D is locked. Move away the hand once the lock sound is heard, then read and record the softness index showed on dial.
5.3.2. Release the upper arm according to process 5.2.2 and takethe specimen away.
Before testing, the suitable ring should be chosen.
•Ring (20mm): to test the footwear leather
•Ring (25mm): to test the sofa leather
•Ring (35mm): to test the cloth or glove leather
The 35mm ring is the test seat itself
If the index is between 0~0.5, the ring should be change
5.4 Test Result: Test three different points of the specimen and take the average value (mm).
1. Clean: Keep clean after eveytime of use with cotton cloth and wipe the probe to keep it clean.
2. Antirust: Spray antirust lubricant to the surface of probe and wipe it clean.
3. Lubrication: Transmission parts should be lubricated every month.
1. For any parts' problem, HS will offer parts / accessories without any charge within 12month.
2. HS technicians can also go to site to repair and maintain the machines freely and the customer just need to pay for the air tickets cost.
3. HS will offer lifetime maintenance.